The measuring system provides the combination of a mechanical tipping bucket rain gauge, a composite sensor (T-P-RH), an ultrasonic anemometer, all third parties, and a remote station completed under the Paradigm.
measuring instruments includes model “15189 produced by lambrecth with tipping bucket in stainless steal WMO standard.Termo hygrometer t026 TTEPRH produced by siam+ micros is a combined sensor for measuring air temperature and relative humidity. Temperature measurement is taken using a platinum Pt100 resistance thermometer with a response curve, compliant with the DIN 43760 Class 1/3 standard and 4-wire connection.The humidity measurement is obtained using a laser-cut capacitive polymer transducer connected to an electronic signal conditioning board.the anemometer LCJ captures CV7-l ultrasonic sonsor measures wind in m/s
the station includes an Electronic board (ACROboard “Station” v04), the latest release. This card is dedicated to the collection functions and data transmission (via GPRS, with internal antenna) the supply management of energy to the components (and charging the buffer battery).
Power is supplied by a 20W Solar Panel, combined with a 12V buffer battery which provides 12 Ah.
|Environment:||rural, open field|
|Temperature Excursion:||-15° + 40°|
|Network coverage:||GPRS mobile network, strong and stable signal|
|Energy source:||Solar panel|
|Battery:||12V e 12Ah|
|Backup energy source:||Battery 2N-AIR|
|Secondary connectivity:||Radio Satellitare a bassa frequenza|
For this configuration theDIY KIT is available
The DIY kit contains all the material and the documentation necessary for the realization in own and on-site installation of your monitoring station ACRONET. End user remains only the task of assembling the individual components and place them in a “work” according to the types of fasteners indicated. For configurations connected to the infrastructure via wireless GPRS simply insert a SIM card enabled for GPRS data traffic (2G). From that moment all data read and sent according to the file will be sent to the infrastructure set-up and remains available to portal ACRONETWORK.ORG.